Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.

Showing results for

- Navigation
- Simcenter
- Forums
- Blogs
- Knowledge Bases

- Siemens PLM Community
- Simcenter
- Testing Forum
- LMS Test.LAB: Acoustic order tracking measurement ...

- Subscribe to RSS Feed
- Mark Topic as New
- Mark Topic as Read
- Float this Topic for Current User
- Bookmark
- Subscribe
- Printer Friendly Page

- Mark as New
- Bookmark
- Subscribe
- Subscribe to RSS Feed
- Permalink
- Email to a Friend
- Report Inappropriate Content

11-26-2015 02:54 AM

Dear all,

I had measured an acoustic order tracking. Then I can create a picture with the acoustic order. In this picture I can calculate on the y-axis under processing a weighting e.g. like A. When I want the same with the overall level of the acoustic channel then I get the message from the program: Weighting can only be applied on frequency spectra.

My question: Why can I calculate only the orders not the the overall level with a weighting?

The weighting with an overall level I can only create with the a measurement under Online Processing. It is not possible, when I forgot a weighting e.g. A at the measurement with an overall level, to make a subsequent calculation in a picture.

Why not?

The only way to calculate it later, is the detour via the Time Data Processing.

Is there an easier way ?

Thank you for your reply

Kurt

Labels:

2 REPLIES

- Mark as New
- Bookmark
- Subscribe
- Subscribe to RSS Feed
- Permalink
- Email to a Friend
- Report Inappropriate Content

12-01-2015 10:55 AM

Hi Kurt - As an overall level function is a calculation of the total energy in the frequency range measured, the underlying frequency content is lost. Therefore, any weighting desired can no longer be applied, as of course the curves for A-, B-, etc. are frequency dependant. As documented in the online help, the Common Overall Level is described as:

*This function is based on the frequency domain computation of the rms of the block being acquired. The rms (root mean square) is computed by squaring values over the frequency range. Then the square root of the mean is taken. So, for a frequency spectrum block, the OA level is given by:*

I'll ask around to see if anyone has suggestions for weighting overall level curves in an easier fashion that recalculating in Time Data Processing.

- Mark as New
- Bookmark
- Subscribe
- Subscribe to RSS Feed
- Permalink
- Email to a Friend
- Report Inappropriate Content

12-02-2015 11:56 AM

Follow Siemens PLM Software

© 2017 Siemens Product Lifecycle Management Software Inc