Turn on suggestions

Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.

Showing results for

- Navigation
- Simcenter
- Forums
- Blogs
- Knowledge Bases

- Siemens PLM Community
- Simcenter
- Testing Knowledge Base
- Calculate a VECTOR SUM to make understanding vibra...

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

Calculate a VECTOR SUM to make understanding vibration data easy!

Siemens Experimenter

07-08-2016
06:00 AM

Use a VECTOR SUM to simplify your data analysis!

Products and humans experience vibration in 3 dimensions/directions. Whether a particular vibration is in the lateral direction or in the vertical direction, it can still fatigue a part or lead to operator discomfort.

A VECTOR SUM can be applied to the three directions of a triaxial accelerometer to calculate the overall magnitude of vibration, independent of the direction.

The formula for a VECTOR SUM of a triaxial vibration measurement consisting of X, Y and Z is as follows:

So for a vibration levels where X=5g, Y=7g and Z=12g, the VECTOR SUM would be equal to: 14.76g as follows:

However, when using Simcenter Testlab (formerly LMS Test.Lab), complex quantities (with both a real and imaginary portion) are often measured. For example, spectrums can include phase. In reality, the VECTOR SUM equation is a bit more complicated, as follows:

**Calculating a VECTOR SUM in Simcenter Testlab Signature**

In Simcenter Testlab Signature, one can calculate a VECTOR SUM either:

- during a measurement with real time viewing OR
- after a measurement has already been performed in throughput processing

To calculate a VECTOR SUM, set it up in either:

**During Measurement:**In “Online Processing worksheet” of Simcenter Testlab Signature, choose “Derived” tab in upper right OR**Offline Processing:**In “Time Data Processing” worksheet, click on the “Change Settings” button from the “Channel Processing” area and select the “Derived FS” tab (Note: "Time Data Processing" can be turned by selecting "Tools -> Add-ins -> Signature Throughput processing" from the main menu)

In the “Derived” tab, highlight the first cell in the “Formula” column and press the “F(x)” button above.

Scroll to the bottom of the “Select Function” menu and choose the VECTOR_SUM function. You may need to click the “New function” button once to make three entries possible in the menu.

In the VECTOR_SUM dialog, enter either the Channel reference (CH1, CH2, CH3, etc) or the PointID (left_mnt_act:+X, left_mnt_act:+Y, etc). If you do not know the Channel reference, it is available in the drop down menu in the upper right of the 'Channel Processing Settings' menu.

Either method (the name or number) can be used to refer to a channel. Press OK when finished.

Now a VECTOR_SUM will be calculated on any and all frequency-based data functions: spectrums, autopowers, orders, frequency sections, etc.

**Conclusion**

Once a VECTOR SUM calculation is setup and created, it is much easier to interpret vibration data. Instead of having to look at three individual graphs for each vibration measurement location, only one graph needs to be viewed to ascertain the overall vibration.

**Simcenter Testlab Acquisition Tips**

- Simcenter Testlab Signature
- Hands-free acquisition (almost)
- Capturing Transient Events in Simcenter Testlab
- Cool Channel Setup tricks for Triaxial Accelerometers
- Data Acquisition Based on GPS Location
- How to calculate a VECTOR SUM
- Tips and tricks for acquiring torsional orders
- CAN Bus Measurements
- No more tach! Use OBDII to get engine and vehicle speed...
- Record Raw CANBUS Data and Decode Offline!
- Single Plane Balancing in Simcenter Testlab
- Measuring strain gauges in Simcenter Testlab
- Correcting a bad calibration factor
- Simcenter Testlab Calibration

**Simcenter Testlab Processing Tips**

Comments

Jeongsoo

Enthusiast

- Mark as Read
- Mark as New
- Bookmark
- Permalink
- Email to a Friend
- Report Inappropriate Content

09-21-2017
12:29 PM

It is quite useful method for me to assess vibration severity.

Many thanks!

Jeongsoo

Rohan_Chaudhari

Experimenter

- Mark as Read
- Mark as New
- Bookmark
- Permalink
- Email to a Friend
- Report Inappropriate Content

04-25-2018
10:34 AM

Kevin_Grenier

Siemens Phenom

- Mark as Read
- Mark as New
- Bookmark
- Permalink
- Email to a Friend
- Report Inappropriate Content

04-26-2018
04:27 AM

Hello, since the VECTORSUM command is a Derived Channel on Online Processing in Signature Acquisition or Channel Processing in Time Data Processing \ Signature Throughput Processing it would be done on the frequency domain data. The screen shows it as a Frequency spectrum so it is being calculated from the spectrums.

I would personally not every try a vector sum on the time domain data. So, to answer your question, it is FFT --> VectorSum(FFT).

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

Related content

Follow Siemens PLM Software

© 2019 Siemens Product Lifecycle Management Software Inc