Turn on suggestions

Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.

Showing results for

- Navigation
- Simcenter
- Forums
- Blogs
- Knowledge Bases

- Siemens PLM Community
- Simcenter
- Testing Knowledge Base
- How to calculate damping from a FRF?

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

How to calculate damping from a FRF?

Siemens Phenom

07-11-2016
10:44 AM

**** Free On-Demand Webinar: Basics of Modal Analysis ****

A classical method of determining the damping at a resonance in a Frequency Response Function (FRF) is to use the “3 dB method” (also called “half power method”).

In a FRF, the damping is proportional to the width of the resonant peak about the peak’s center frequency. By looking at three dB down from the peak level, one can determine the associated damping.

The “quality factor” (also known as “damping factor”) or “Q” is found by the equation Q = f0/(f2-f1), where:

- f0 = frequency of resonant peak in Hertz
- f2 = frequency value, in Hertz, 3 dB down from peak value, higher than f0
- f1 = frequency value, in Hertz, 3 dB down from peak value, lower than f0

**What is damping?**

Damping is the energy dissipation properties of a material or system under cyclic stress.

If a mechanical system had no damping, once set in motion, it would remain in motion forever. Damping causes the system to gradually stop moving over time. The more damping present in a mechanical system, the shorter the time to stop moving.

**Forms of Damping**

Damping can be expressed in several different forms, including “loss factor”, “damping factor”, “percent critical damping”, “quality factor”, etc. Note that if the value of one of these damping forms is known, the other forms can be mathematically derived. It is just a matter of using the equations to transform the value to a different form.

For the purposes of this article, we will consider the “damping factor”, “quality factor” and “Q” to be the same as described below:

Why have different ways of expressing damping? Mostly, this is to make it easier to discuss differences in damping.

For example, perhaps two materials were tested, and the “percent critical damping” was 0.0123% in one case, and 0.0032% in the other case, which sounds like a small difference. In terms of “quality factor” or “damping factor”, the difference is 4065 versus 15625, which sounds much bigger!

Instead of using long numbers with small decimal differences like percent critical damping, changing to the “quality factor” form makes the differences easier to understand.

**More vs Less Damping**

As the peaks in a FRF get wider relative to the peak, the damping increases (i.e., “more” damping). This means that any vibration set in motion in the structure would decay *faster* due to the increased damping.

Depending on the form being used to express damping, the value may be higher or lower.

For example, the “quality factor” or “damping factor” will decrease with more damping, while “loss factor” and “percent critical damping” would increase with more damping.

**Cautions**

When calculating damping from a FRF, there are several items which can influence the final results:

**Caution #1: Frequency Resolution**

When using a digital FRF, the data curve is not continuous. It is broken into discrete data points at a fixed frequency interval or resolution. For example, this could be a 1.0 Hz spacing versus a 0.5 Hz spacing between data points. This will influence how the frequency values f0, f1 and f2 are determined.

Using a very fine frequency resolution is recommended when calculating damping using the 3dB method.

**Caution #2: Window**

Sometimes, when using a modal impact hammer to calculate an experimental FRF, a user may apply an Exponential Window to avoid leakage effects to the accelerometer signal. The accelerometer response is multiplied the exponential window causing it to decay quicker in time, thus increasing the apparent damping.

One should be aware the resulting FRF will yield higher damping estimates when using the “3 dB method”. It is possible to back out the effects of the window and get the actual damping value. For example, in Simcenter Testlab:

- Modal Curvefitting - When performing a modal analysis curvefit, the exponential window affect is removed from the modal damping estimate automatically for the user.
- Damping Cursor in Display - With Simcenter Testlab Revision 18 and higher, the damping value reported by the cursor automatically adjusts for the effects of an exponential applied to the data.

**Caution #3: Mode Spacing**

If two modes are close to each other in frequency, it may be impossible to use the “3dB method” to determine the damping values.

If this is the case, there are two resonant peaks close together, which would make the peak wider than if each peak could be analyzed separately. In this case, one should use a modal curvefitter to determine the damping properly for each peak. A modal curvefitter can successfully separate the two modes influence on each other.

**Simcenter Testlab Damping Cursor**

After displaying a FRF, to calculate damping in Simcenter Testlab (formerly called LMS Test.Lab):

- Right click and add an “Automatic -> Peak Cursor” (or a regular SingleX cursor if one is confident in positioning it on the actual peak). Note: It is not necessary for amplitude to be in dB of the FRF, this will be accounted for correctly.

- After the cursor is on the peak value, right click on the cursor and select “Calculations -> Q” or “Calculations -> Damping Ratio” as desired

- Right click on cursor and select “Automatic Peak Parameters” and add as many peaks as desired in the “Max number of extrema” field.

- Right click on legend and choose “Copy Values” to export the values to Excel if desired

Have more questions on using the half power method to calculate damping from a FRF? Feel free to email

peter.schaldenbrand@siemens.com or contact Siemens PLM GTAC support.

**More Structural Dynamics Links:**

- Free On-Demand Webinar: The Basics and Math behind Modal Analysis
- Natural Frequency and Resonance
- What is Frequency Response Function (FRF)?
- The FRF and it Many Forms!
- What modal impact hammer tip should I use?
- Simcenter Testlab Impact Testing Procedure
- Getting Started with Modal Curvefitting
- What is an Operational Deflection Shape (ODS)?
- Roving accelerometer versus roving hammer modal analysis test
- Simcenter Testlab setup tricks for triaxial accelerometers
- Modal Assurance Criterion (MAC)
- Frequency resolution, Sampling frequency, Spectral lines, and more...
- Alias Table: Mapping Test Data to Geometry
- Ground Vibration Testing and Flutter

**Simcenter Testlab (formerly LMS Test.Lab) Display tips:**

- Simcenter Testlab: Display Layouts
- Simcenter Testlab: Curve Linestyles
- Simcenter Testlab: Cursor Tips and Tricks
- Simcenter Testlab: Coupled Cursors and Coupled Limits
- Simcenter Testlab: Interactive order cursor!
- Simcenter Testlab: Order Cursor vs Time in Colormap
- Simcenter Testlab: The "Double X cursor"
- Siimcenter Testlab: Automatic Peak-Valley Cursor
- Simcenter Testlab: Harmonic Cursors
- Simcenter Testlab: XY Graph
- Simcenter Testlab: Keyboard and Mouse Shortcuts
- Simcenter Testlab: Matrix-Heatmap Display

**Other Articles:**

Labels:

Comments

Alvaro12

Creator

- Mark as Read
- Mark as New
- Bookmark
- Permalink
- Email to a Friend
- Report Inappropriate Content

10-03-2018
01:20 AM

Good night friends.

Interesting Topic.

I'd like to know a little more about this topic.

- For example an introduction to use this dynamic property in the design by finite elements.

- Some practical aplication of this value, independent from the finite model.

- For example, the natural frequency have a important use to avoid resonance condition. About this value, without relation to the optimization design, I'd like to know practical utility.

I'd like to know a little more about this topic.

Thanks in advance

Alvaro

JeeryGu

Enthusiast

- Mark as Read
- Mark as New
- Bookmark
- Permalink
- Email to a Friend
- Report Inappropriate Content

12-23-2018
10:23 PM

Thanks for introducing about damping ratio clearly

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

Related content

Latest articles

Subject | Date Posted | |
---|---|---|

Follow Siemens PLM Software

© 2019 Siemens Product Lifecycle Management Software Inc