Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.

Showing results for

- Navigation
- Simcenter
- Forums
- Blogs
- Knowledge Bases

- Siemens PLM Community
- Simcenter
- Testing Knowledge Base
- LMS Test.Lab: The Versatile "Double X" Cursor

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

Siemens Genius

08-06-2016
06:29 PM

**LMS Test.Lab: The Versatile "Double X" Cursor**

In LMS Test.Lab, the Double X cursor is a wonderful and powerful cursor for data analysis.

**Double X: The Basics**

To add a Double X cursor to a display, right click and choose “Add Double Cursor -> X” (Hint: Hold down the CTRL key when right clicking on the display to always get the cursor menu)

On the Double Cursor, the left and right cursors can be moved (*Picture 2*):

- Left Cursor: Moves both cursors together at the same time. The cursors stay fixed distance apart
- Right Cursor: Moves only the right cursor, changes the distance between the two cursors

Set the cursor to the frequency range desired to perform an analysis.

**Double X: Max**

Need the maximum over a frequency/rpm range? Want to know the frequency/rpm the max occurs at?

Right click on either the left or right of the Double Cursor and select “Calculations -> Max” (*Picture 4*).

The maximum appears in cursor legend, along with the frequency at which the maximum value occurs (*Picture 5*).

Right click on the cursor legend and choose "Copy Values" to paste the maximum value into Excel.

**Double X: Frequency Calculator**

Looking at some time domain data and wondering what the frequency is present (*Picture 6*)?

Right click on the Double X cursor and select “Calculations -> Freq (1/deltaT)” as in *Picture 7*.

**Double X: Linear Regression**

Trying to figure out your average acceleration from a RPM vs time trace (*Picture 8*)?

To determine the rate of speed increase (ie, the acceleration) from the RPM vs time, right click on the cursor and choose "Calculations -> Linear Regression" (*Picture 9*).

The "Linear regression" function fits a line to the data between cursors. The slope and offset of this line are displayed in the legend.

Right click on the legend and choose "Copy Values" to copy/paste the slope and offset to Excel.

**Double X: RMS and Color**

Want to know the RMS level over a specific frequency range?

Right click on the Double X cursor and choose “Calculations -> RMS” as shown in *Picture 11*.

Want to highlight the frequency range of interest with a little pizazz? Add a bit of color by right clicking on the Double X cursor and selecting “Properties”.

Turn on “Fill with color:” (*Picture 12*) and select some beautiful colors (*Picture 13*)!

Even mix colors with overlapping frequency ranges!

**More LMS Test.Lab Display Tips:**

- LMS Test.Lab from Siemens PLM
- LMS Test.Lab Displays: Layouts
- LMS Test.Lab Displays: Curve Linestyles
- LMS Test.Lab Displays: Cursor Tips and Tricks
- LMS Test.Lab Displays: Coupled Cursors and Coupled Limits
- LMS Test.Lab Displays: Interactive order cursor!
- LMS Test.Lab Displays: The "Double X cursor"
- LMS Test.Lab Displays: Harmonic Cursors
- LMS Test.Lab Displays: XY Graph
- LMS Test.Lab Displays: Keyboard and Mouse Shortcuts
- RMS Calculations and Overall Level

Labels:

Comments

Kevin_Grenier

Siemens Legend

- Mark as Read
- Mark as New
- Bookmark
- Permalink
- Email to a Friend
- Report Inappropriate Content

08-08-2016
09:09 AM

Great article! I've noticed sometimes that the double X cursor calculations do not work for large time histories. When that happens the time data icon is half blue and half red, rather than all blue.

This indicates that more data points are available than the option "Max. number of samples used for calculations. To adjust: 1) Go to Tools, Options, Data and increase the number shown for "Max. number of samples used for calculations" by adding one or two zeros to the number. By default, one million data samples are used. When displaying more than one million data samples the icon will be half blue and half red, and the calculations will not be done unless you move the double X cursors to inlcude less than than one million data samples. Press Apply or OK. 2) Refresh the few on the Navigator and the icon should be all blue and your calculations will now work.

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

Related content

Follow Siemens PLM Software

© 2017 Siemens Product Lifecycle Management Software Inc