turn on suggestions

Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.

Showing results for

- Navigation
- Simcenter
- Forums
- Blogs
- Knowledge Bases

- Siemens PLM Community
- Simcenter
- Testing Knowledge Base
- Root Mean Square (RMS) and Overall Level

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

Siemens Experimenter

a week ago

There are several different ways that the term Root Mean Square (RMS) is used.

This article will go over the following:

**RMS Amplitude Format:**How the equivalent energy of a sine wave is expressed in a spectrum**RMS of a Spectrum:**A representation of the overall energy in a spectrum – often called the overall level**Tracked RMS / Tracked Overall Level:**Tracking the RMS of each average against a tracking parameter to see how the energy content changes**RMS in LMS Test.Lab**

**RMS Amplitude Format:**

Amplitude formats describe how the amplitude of a sine wave is expressed in a spectrum.

The RMS amplitude format is valuable because it indicates the equivalent steady state energy value of an oscillating signal.

For example, a sine wave (like a pressure wave) oscillates above and below zero Pascal. The average value of the wave is zero. This is not a good representation of the effective energy of the wave. To get a better understanding of the equivalent steady state value, the RMS is used.

The *RMS amplitude format *is calculated by squaring the peak amplitude (A) of the sine wave, diving it by two, and then taking the square root of that quantity. For a single sine wave, the RMS amplitude can be represented as 0.707*A.

Look at the spectrum below. The curves are the *exact same data* but expressed with different amplitude formats. The black curve is expressed in peak format. The pink curve is expressed in RMS format. Notice that all amplitude levels of the pink curve are below the black curve (Figure 2).

In the zoomed in section, each spectral line is represented by a cross. When converting to RMS amplitude format, each spectral line is multiplied by ~0.707. For example, the spectral line at 694Hz has a peak amplitude of 1.000g. This corresponds to a RMS amplitude of 0.707g.

The RMS amplitude format is used to represent the equivalent steady state value of a sine wave at each spectral line.

**RMS of a Spectrum:**

Often, an RMS value of a spectrum is desired to be calculated. The RMS of a spectrum is a single number that represents the overall level of energy across a frequency range.

In the graphic below, the RMS of the spectrum is 54.08g.

To calculate the RMS of a spectrum, the root sum square of all the spectral lines within the frequency range of interest must be calculated.

The “A” term in the above equation represents the amplitude of a spectral line. It is important that this amplitude (A) value is in RMS format and has window correction factors taken into consideration (more about this below).

A_0 is the first spectral line under consideration, A_k is the last spectral line under consideration.

There are some important considerations that must be taken to ensure the calculated RMS value is correct:

1. The individual spectral line values (A) must be in *Linear* *format*. If the units were squared or power units, they must be squared rooted before performing the calculation. For example, an autopower of sound pressure with units of Pa2 must be square rooted to units of Pa.

2. Each individual spectral line must be in *RMS format* for the calculation. No matter what format is used for *displaying* the spectrum, RMS amplitude format is used when *calculating* the RMS of the spectrum. The RMS calculation produces the same result no matter how the spectrum is displayed.

3. Window correction factors may need to be applied to the spectrum. If a window was used during acquisition, energy correction must be applied to the spectrum. The usage of windows to compensate for leakage can cause spectral lines to appear lower in amplitude / energy content than they really are. To get appropriate values for RMS spectral energy, energy correction must be applied.

The RMS calculation has a pre-requisite of linear, RMS, energy corrected values.

The RMS value calculated in LMS Test.Lab will automatically convert data to energy, RMS, and linear values "behind the scenes", regardless of how it is displayed in the software interface. This ensures consistent values for RMS calculations.

Users who want to manually calculate RMS must take these elements into consideration.

**Tracked Overall Level:**

The RMS value of a spectrum is often called the overall level. The overall level can be tracked versus speed or time to see how the amount of energy in the signal changes.

When recording an event that is changing, it is often preferred to track the event on time or RPM. This means that for every increment of time or RPM a spectrum is calculated.

In the below example, a spectrum is calculated every 25 rpm between 3500 RPM and 4000 RPM. The overall level (RMS) for each of these spectrums is calculated.

Then, these RMS values are plotted vs. RPM. This way, it is possible to see which RPM values have especially high (or higher than expected) energy content.

**RMS in LMS Test.Lab:**

This section will cover:

- Changing Amplitude Format in LMS Test.Lab
- Calculating the RMS of a frequency spectrum
- Calculating the RMS over a specific frequency range
- Calculating Tracked Overall Level in Test.Lab

*Changing Amplitude Format in LMS Test.Lab:*

To change the amplitude format in LMS Test.Lab, right lick on the y-axis of the plot, select “Processing” and then choose an amplitude format under “Spectrum & Section Scaling”.

*Calculating the RMS of a frequency spectrum:*

To calculate the RMS of a spectrum in LMS Test.Lab, right click on the legend and choose “Options”.

Next, calculate the RMS of the spectrum by selecting RMS in the “Calculated Content” tab, and the clicking the “Add to selection” arrow. Additionally, the Unit Label can be turned on in the lower right.

The RMS value of the entire spectrum will now be displayed in the legend. Amplitude format and energy corrections are applied in the background as needed.

*Calculating the RMS over a specific frequency range:*

Alternatively, the RMS value over a smaller frequency range can also be calculated.

Add a double cursor to the plot and position the cursor over a range of values.

Then, right click on the cursor, and select Calculations -> RMS.

The RMS value of the content between the cursors will be calculated and displayed in the cursor legend.

*Calculating Tracked Overall Level in Test.Lab:*

To calculate a tracked overall level in LMS Test.Lab, ensure the measurement mode is set to “Tracked”.

In the Section Settings dialog, make sure that “Overall level” is checked on under the “Overall Level” tab.

The data will be saved in the Overall Level folder within the Sections folder.

**Conclusion: **

It is important to understand the difference between RMS amplitude format, the RMS of a spectrum, and the tracked RMS.

RMS amplitude format represents the equivalent steady state value of a sine wave in a spectrum.

The RMS value of an entire spectrum represents the overall energy level. This value can be tracked to better understand how it changes.

**Questions?** Contact us.

**Related Links**

- Siemens LMS Test.Lab Product Information
- Digital Signal Processing: Sampling Rates, Bandwidth, Spectral Lines, and more...
- Gain, Range, Quantization
- Aliasing
- Overloads
- Averaging Types: What's the difference?
- Spectrum versus Autopower
- Autopower Function...Demystified!
- Power Spectral Density
- Windows and Leakage
- Window correction factors
- LMS Test.Lab: The Versatile “Double X” Cursor

Labels:

- Article History
- Subscribe to RSS Feed
- Mark as New
- Mark as Read
- Bookmark
- Subscribe
- Email to a Friend
- Printer Friendly Page
- Report Inappropriate Content

Related content

Latest articles

Follow Siemens PLM Software

© 2017 Siemens Product Lifecycle Management Software Inc